Stainless steel test screen for laboratory screening
|Model No.: ||CW|
|Min. Order: ||1 Set/Sets|
|Payment Type: ||L/C,T/T|
|Packaging: ||Plywood case or others per customer's request|
|Delivery Time: ||15 Days|
|Place of Origin: ||Henan, China (Mainland)|
|Supply Ability: ||15 sets of 20 days|
|Certificate: ||CE ISO SGS|
The experimental sieve is a standard experimental sieve which integrates powder grading and liquid filtration; it is selected for laboratory and quality inspection room; it has accurate particle size structure analysis of particles and powders, and has the characteristics of low noise, high efficiency and high precision. Special power generator, generate multi-frequency vibration, high-efficiency sieve sample; standard sieve body to ensure high precision of sample analysis; electronic delay device, can set the same material screening time according to needs, to ensure the uniformity of sample analysis data , to minimize the test error, in order to make an accurate judgment on product quality.
Test sieve operation method
1. The test sieve should be placed horizontally, confirm that the power supply and nameplate requirements are consistent, and ensure that the grounding; the vibration part can not be in contact with other objects; whether the bolts of each part are locked.
2. Determine the standard sieve frame to be used according to the materials to be inspected and the corresponding standards.
3. Stack the standard sieve frames from small to large, and stack them from bottom to top on the base of the screen frame.
4. Put the material to be inspected into the uppermost standard sieve frame, cover the standard sieve upper cover, adjust the height of the adjustment rod according to the total height of the standard sieve frame, and then press and clamp the standard sieve frame with the pressure plate and the lock nut ( Note: The force on both sides should be consistent).
5. According to the nature of the material and the amount of material to be fed, set the running time on the timer, then turn on the power switch, and the standard test sieve will start working.
6. After the standard test sieve is stopped, unscrew the lock nut, remove the pressure plate, and carefully remove the standard sieve frame.
7. Turn off the power.
CW test sieves are widely used in inspection departments such as laboratories and quality inspection rooms. The products are characterized by compact structure, low noise, high efficiency and extremely accurate. The high-quality new power generator is used to generate multi-frequency high-frequency vibration, and the materials are efficiently screened and measured. At the same time, according to different materials, different working hours can be realized by electronic delay device, the inspection error can be minimized, the consistency of sample analysis data can be ensured, and the product quality can be accurately judged. It is mainly used to detect the particle size distribution of granular materials, or the content of impurities in products, and the content of solids in liquid materials.
1. You can check our products on Alibaba.com before you sell. If you have any questions, please feel free to contact us.
2. We provide detailed video teaching installation during sales.
3. We have a one-year warranty after the sale